- Show/Hide Grid Ctrl+Shift+G
- Toggle the grid on/off by using this menu entry or the associated
- Frequency Axis Log/Linear Ctrl+Shift+A
- Toggle the frequency axis between logarithmic and linear display.
The button on the toolbar provides the same function.
- Undo Zoom Ctrl+Z
- Restore the graph axis settings that were in use when the middle or right
mouse button was last pressed with the pointer in the graph area. The
main application is to undo the last
Variable Zoom or
Zoom to Area action, but
it can also be used to undo a pan or any other action that altered the
axis settings since the right mouse button was last pressed in the graph
- Fit graph to data Ctrl+Alt+F
- Set the graph axes to show the full span of the data. The Fit to data
button on the graph limits dialog has the same effect.
- 1/1 Octave Smoothing Ctrl+Shift+1
- 1/2 Octave Smoothing Ctrl+Shift+2
- 1/3 Octave Smoothing Ctrl+Shift+3
- 1/6 Octave Smoothing Ctrl+Shift+6
- 1/12 Octave Smoothing Ctrl+Shift+7
- 1/24 Octave Smoothing Ctrl+Shift+8
- 1/48 Octave Smoothing Ctrl+Shift+9
- Variable Smoothing Ctrl+Shift+X
- Psychoacoustic Smoothing Ctrl+Shift+Y
- ERB Smoothing Ctrl+Shift+Z
- Apply a smoothing filter to the current channel. Repeating the action
removes the smoothing. Variable smoothing applies 1/48 octave below 100 Hz,
1/3 octave above 10 kHz and varies between 1/48 and 1/3 octave from
100 Hz to 10 kHz, reaching 1/6 octave at 1 kHz. Variable smoothing is recommended
for responses that are to be equalised.
Psychoacoustic smoothing uses 1/3 octave below 100Hz, 1/6 octave above 1 kHz and
varies from 1/3 octave to 1/6 octave between 100 Hz and 1 kHz. It also applies more
weighting to peaks by using a cubic mean (cube root of the average of the cubed values)
to produce a plot that more closely corresponds to the perceived frequency response.
ERB smoothing uses a variable smoothing bandwidth that corresponds to the ear's
Equivalent Rectangular Bandwidth, which is (107.77f + 24.673) Hz, where f is in kHz.
At low frequencies this gives heavy smoothing, about 1 octave at 50Hz, 1/2 octave at 100 Hz,
1/3 octave at 200 Hz then levelling out to approximately 1/6 octave above 1 kHz.
- Remove Smoothing for Current Measurement Ctrl+0
- Set smoothing to None. Pressing the smoothing shortcut again will also remove
smoothing - for example, pressing Ctrl+Shift+3 will smooth the data to 1/3
octave, pressing it again will remove the smoothing.