The All SPL graph is an overlay graph that shows all measurements (SPL and/or
Impedance) that have been made. It allows an average to be generated of all selected
traces or arithmetic operations to be carried out on pairs of traces to generate
a new trace.
Average The Responses calculates an rms (root mean square) average of the
SPL values of those traces which are selected when the button is pressed. That means
the dB values are converted to linear magnitudes, those magnitudes are then squared,
summed and divided by the number of measurements, the square root of the result is
taken, then the value is converted back to dB. Phase is not taken into account,
measurements are treated as incoherent. The frequency range of the averaging result
covers the region where the traces overlap, for example if one trace was measured
to 200Hz, another to 500Hz and a third to 1000Hz the average would range to 200Hz (to
the lowest end frequency). If only a single trace is selected the result has the magnitude
data from the source measurement and no phase data.
The control panel for the All SPL graph has these controls:
The controls are split into four groups:
- Those which apply to an individual measurement, shown by the Measurement actions button
- Alignment tool controls, shown by the Alignment tool button
- Trace Arithmetic controls, shown by the Trace arithmetic button
- Those which apply to all measurements currently selected in the graph legend
The Actions for all selected measurements are:
- A smoothing control that allows the fractional octave smoothing
setting for all the currently selected traces to be changed. The
chosen setting is applied using the Apply smoothing button
- Time align, which brings all the currently selected measurements
into time alignment. If the measurements have been made with a timing
reference (a loopback connection or an acoustic timing reference) the
impulse response is shifted according to the measurement delay value,
taking into account any IR timing offsets which have been applied
since the measurement delay was calculated. Measurements which have been
made without a timing reference are shifted according to the estimated IR
delay. Time alignment can only be applied to measurements that have an impulse
- Align IR start, which aligns all the currently selected measurements
using their IR start times.
- Align SPL..., which adjusts all the currently selected SPL measurements
so that they have the same average SPL over a selected span.
- dB average, which averages the dB SPL values of the currently selected traces.
This produces results that are closer to what one might intuitively expect from
looking at the traces, but gives equal weight to peaks and dips which masks the magnitude
difference between them. For example, a +20 dB peak has a magnitude that is 100 times
larger than a -20 dB dip. dB averaging them produces 0 dB, 10 times smaller than the
peak and 10 times larger than the dip. dB averaging may be useful when averaging
smoothed traces to derive an EQ target, with unsmoothed data the dips would have
a disproportionate effect on the result.
- Vector average, which averages the currently selected traces taking into
account both magnitude and phase. It can only be applied to measurements that
have an impulse response and is most appropriate for multiple measurements taken
from the same position, or measurements which have been time and level aligned
before averaging them.
- RMS average, which calculates an rms average of the SPL values of
those traces which are selected when the button is pressed. This does the same
as the Average The Responses button, the dB values are converted to linear
magnitudes, those magnitudes are then squared, summed and divided by the number
of measurements, the square root of the result is taken, then the value is converted
back to dB. Phase is not taken into account, measurements are treated as incoherent.
If only a single trace is selected the result has the magnitude data from the source
measurement and no phase data. If the measurements were made at different positions
(spatial averaging) it is usually best to first use the Align SPL... feature
to remove overall level differences due to different source distances.
- Save to file, which saves the currently selected measurements to an mdat file.
- Export as txt, which exports the currently selected measurements to text files.
- Export IR WAV, which saves the impulse responses of the currently selected measurements to WAV files.
- Export IR txt, which saves the impulse responses of the currently selected measurements to text files.
The Measurement actions dialog has the following controls:
The options for each selected measurement are to:
- Offset the measurement either temporarily or (by using
Add offset to data) permanently
- Make a Minimum phase version of the selected measurement to use in
calculations - the copy has a minimum phase impulse response and the
same magnitude response and calibration data as the original
- Make an Excess phase version of the measurement to use in
calculations - the copy is the original measurement divided by a minimum
phase version of the measurement
- Make a new measurement with Mic in box correction that has
the SPL data of the source with a correction to compensate for the
measurement being made using the microphone in box method, adding an offset
of 40*log10(f/f0) where f0 is a nominal reference frequency, which has
been set to 50 Hz. The result would usually be merged with a far field
- Make a new measurement with Response copy that has the same
response as the measurement - the copy has the same impulse
response and calibration data as the original but does not include
distortion data or equaliser settings, for example
The Trace Arithmetic controls allow the chosen pair of traces to be
added, subtracted, multiplied, divided, coherently (aka vector) averaged or merged.
If both the chosen traces have impulse responses, the result will also have an impulse response,
however the sample rates must be either the same or related by an integer. For example,
traces at 44.1kHz and 11.025kHz can be combined via an arithmetic operation, the result
will have the higher of the two rates. This allows operations on band limited measurements
which may have been decimated to a lower sample rate.
If the traces have incompatible sample rates, or either does not have an impulse response,
the result will not have an impulse response, but it may have both magnitude and phase
data if both the traces it was applied to had magnitude and phase data, otherwise the
result will only have magnitude data and the traces will be treated as incoherent.
The frequency span of the result of an arithmetic operation will be from the lowest
start frequency to the highest end frequency of the traces operated on. Outside their
frequency range traces are treated as being zero valued, with the exception of the divisor
in a division operation which is treated as being unity outside its range. If the
measurements actually have significant levels outside the measurement range the zero
setting will generate oscillations in frequency and time domains, for best results use
traces that span the full frequency range.
Trace arithmetic notes
- For meaningful results measurements that have impulse responses or phase data should
be properly time aligned before they are combined. An exception is the Merge operation,
for which REW will automatically align both magnitude and phase at the merge frequency,
adjusting the trace B time delay as required for the phase match. The amounts of the
adjustments are shown in the notes of the newly generated measurement.
- The currently applied impulse response window settings are used for each trace. The
result uses the same window settings as trace A unless the operation was Merge B to A,
in which case the window settings for trace B (the low frequency portion) are used. The
window reference time for the result will be at the peak of the result IR. The
left window width will be increased if necessary so that it is at least twice the span
from the start of the result IR to its peak to avoid truncation artefacts in the result.
Any frequency-dependent settings are excluded, applying an FDW to the result would amount
to applying the window twice, as it is already applied to the data used to produce the result.
- The result of arithmetic on measurements that have compatible impulse responses is
smoothed using the measurement A smoothing, unsmoothed data is used during the calculations.
Other measurements use whatever smoothing they already had applied during the calculations
and the result is treated as unsmoothed (or 1/48 octave smoothed if data is 96 PPO).
The Alignment tool allows gain, polarity and delay adjustments to be made to a
pair of measurements with a live preview of the summed result updated as the parameters are
changed (the preview uses log spaced data at 96 PPO so may differ slightly from the final
When Show phase traces is selected the All SPL graph is split with the SPL
traces at the top and phase traces for the two selected measurements and the aligned sum at
The Aligned copy buttons generate a copy of the selected measurement with the
alignment gain, delay and polarity selections applied. Aligned sum generates
a new measurement with the summed results of the alignment settings.
The Level phase at cursor button calculates the delay required to bring
the first measurement phase trace to be approximately horizontal around the frequency
of the cursor in the phase trace graph. Both measurements have the same delay applied
so the relative delays required to align them are not affected. Shifting the phase traces
in that way makes it easier to align them. The delay is only used to update the phase
traces, it is not applied when using Aligned copy or Aligned sum.
Undo levelling removes the levelling delay from both measurements.
The Align phase slopes at cursor button calculates the measurement 2
delay required to give both phase traces approximately the same slope around the
frequency of the cursor in the phase trace graph, so they have approximately the
same group delay. Note that the phase traces may still be far apart, particularly
if one measurement is out of phase with the other. In that case inverting one of
the measurements would bring the phase traces closer together.
The Align phase at cursor button calculates the measurement 2 delay
required to bring both phase traces as close as possible to each other around the
frequency of the cursor in the phase trace graph. Note that a better overall
alignment might be achieved by inverting one of the measurements and then aligning
phase, and/or by manually adjusting the delay.
The Trace options button brings up a dialog that allows the line
type of the graph traces to be changed.
If Show points when zoomed in is selected the individual points
that make up the SPL and phase responses are shown on the graph when the zoom
level is high enough for them to be distinguished (which may only be over part
of the plot)
If Show frequency bands is selected the audio frequency bands
are shown in a stripe above the graph. The bands are:
- Infrasound: below 20 Hz
- Sub bass: 20 to 60 Hz
- Bass: 60 to 250 Hz
- Low midrange: 250 to 500 Hz
- Midrange: 500 Hz to 2 kHz
- Upper midrange: 2 to 4 kHz
- Presence: 4 to 6 kHz
- Brilliance: 6 to 20 kHz
If Show modal frequencies is selected the theoretical modal
frequencies for the room dimensions entered in the
Modal Analysis section of the
EQ Window for the currently selected measurement are plotted at the bottom of the graph.
When impedance is being plotted the axis has a span from 0 to 1 kohm. If
a larger impedance range is required the axis can be switched to logarithmic
with a range up to 1 Mohm using the Use a log axis for impedance
check box. If selected a log axis will be used wherever impedance is plotted.