The Filtered IR graph allow octave and one-third octave filters to be applied to the measurement. It is primarily aimed at examining decay behaviour in different frequency bands and analysing the results per ISO 3382. In addition to the filtered impulse response itself this graph includes traces of the impulse response envelope (ETC) and the Schroeder integral.
The Schroeder Integral is a curve obtained by backwards integration of the squared impulse response, ideally starting from a point where the response falls into the noise and applying a correction (a starting value for the integral) which assumes the rate at which the Schroeder curve is falling continues for the whole response. REW uses an iterative procedure to estimate the best starting point for the integration, often called "Lundeby's Method" (from the paper by A. Lundeby, T. E. Vigran, H. Bietz, and M. Vorländer, “Uncertainties of Measurements in Room Acoustics,” Acustica, vol. 81, pp. 344–355 (1995)). The slope of this curve is used to measure how fast the impulse response is decaying, deriving a figure for "RT60" which is the time it would take sound to decay by 60dB. The curve shown on the Impulse graph is for the currently applied filter, if any. When calculating decay data for the octave and one-third octave RT60 results the impulse is first filtered to the corresponding bandwidth and centre frequency before the Schroeder Integral for that band is determined and the various RT60 measures calculated.
The control panel for the Filtered IR graph has these controls:
Octave and 1/3 octave filters can be selected using the 1/1 and 1/3 boxes, which enable the corresponding set of controls. The required filter frequency can be chosen directly from the drop-down list or stepped through using the buttons either side. The filter is applied to the Impulse Response upon selection, a label in the bottom left corner of the graph shows the current filter setting. The filter remains active until "No Filter" is selected or the boxes are unchecked. The measurement name on all graphs is shown with an indication of the applied filter, for example "Auditorium [250Hz 1/3]".
The impulse response may be plotted with or without normalisation to its peak value according to the setting of the Plot responses normalised control. When normalised plotting is selected the peak will be at 100% or 0 dBFS.
The Time reversed filtering control applies the octave band filters backwards in time, this reduces the filter's own contribution to the measured decay. When using 1/3 octave filters at low frequencies the filter decay time can be significant, over 200 ms for a 100Hz 1/3 filter, for example. Applying the filter in reverse reduces this decay to less than 50 ms, but it does affect the response somewhat, such that Early Decay Time (EDT) figures using Time-Reversed filters may not be valid. Note that this control is independent of and does not affect the time-reversed filtering control on the RT60 graph, but it is initialised to the setting last used for RT60.
Zero phase filtering applies 4th order bandpass filters in two passes through the data, one forwards and one reversed, to give a response with overall zero phase shift. This may be useful when comparing the locations of reflection peaks in filtered impulse responses or the ETCs derived from them as the peaks will not be shifted in time by the effect of the filter's phase response. Note that this control is independent of and does not affect the zero phase filtering control on the RT60 graph, but it is initialised to the setting last used for RT60.
The Show data panel control shows a panel on the graph containing the results
for the decay values. The RT60 figures include the decay range over which they have been
calculated and an "r" value, the regression coefficient, which measures how well the data
corresponds to a straight line. A value of -1 would indicate a perfect fit, values lower
in magnitude than -0.98 indicate the corresponding decay figure may not be reliable.
Unreliable figures are italicised and shown orange.
The parameters available are:
The graph can also show the "Regression Line", which is a line obtained by carrying out least squares linear regression on the Schroeder curve over the range applicable to any particular decay parameter. The selector for which regression line is to be shown is next to the Show Regression Line check box.
ETC Smoothing is used to smooth the envelope (ETC) trace using a moving average filter of the duration specified in the spinner.